- The interferential stimulator (IF) device is a battery operated device that includes two controllable output channels.
- Amplitude, mode and duration of the electrical impulses can be altered with the buttons or knobs.
- The controls are protected by a cover in order to prevent accidental changes to the settings.
- Slide the cover down to make adjustments to the controls.
|Channel||Dual, isolated between channels|
|Weight||Approximately 150 grams, battery included|
|Amplitude||Adjustable 0-60mA peak into 500 ohm load each channel,constant current|
|Carrier Frequency||4000Hz, Fixed (CH1)|
|Modulating Frequency||4-160bps, Adjustable (CH2)|
|IF Frequency Modes||9 modes, 1 constant mode, 5 frequency shift modes, 3 auto sweep Modes|
|Constant Mode||4-160bps, Adjustable|
|Auto Sweep||4-45 bps|
|Frequency Shift||Five modes,|
1/1 abrupt shift
6/6 abrupt shift
10/10 abrupt shift
|Sweep Time||15 seconds|
|Frequency Shift Percent||Frequency shifts from 30% below the set frequency to 60% above and then return to 30% below the set frequency.|
|Output Configuration||Quad polar (4 electrodes)|
|Wave form||Symmetrical Balanced Sine Wave|
|4-160 bps, Adjustable, 4 bps/step|
|Pulse Duration||125 µs maximum|
|Patient Compliance Meter||Display total treatment time|
|Lock||Prevents the patient from accidentally changing any of the set parameters|
|Timer||15, 30 minute selectable|
|LCD||Displays mode, bps rate, abrupt/ramp, timer, and channels|
|Max charge per Pulse||15 Micro-coulombs Maximum|
|Low Battery||A low battery indicator will show when the battery is low.|
|Power Source||9 Volt DC adapter, 500mA or 9 volt alkaline battery (not recommend due its short life)|
|Tolerance||There may be +/-10% tolerance of settings and 20% tolerance of output.|